Thorough cleaning of medical instruments and endoscopes is a basic requirement for effective disinfection. The well-chosen combination of four enzymes with surfactants and other ingredients of Sekusept Multienzyme P offers you an excellent cleaning performance. Sekusept Multienzyme P is the ideal enzymatic cleaner for all types of medical instruments, including flexible endoscopes.
|Our performance||Your advantage|
|Innovative combination of four enzymes with powerful surfactants||Strong, self-cleaning Reduces pre- and post-cleaning expenses RKI-conforming pre-cleaning|
|Comprehensive material compatibility||Suitable for pre-cleaning of thermo unstable and thermo stable medical instruments and flexible endoscopes|
|2 L bottles with viewing window||Optimum fill level inspection and secure dosing|
|Low foam||Suitable for ultrasonic application|
|Transparent blue coloring||Visible cleaning solution with good recognition of the instruments|
Depending on the type and degree of stains, prepare a solution of 0.3% (3ml/l) to 2% (20ml/l). Place the instruments to be cleaned in the solution, ensuring that all surfaces are completely wetted. Clean the instruments paying attention to the manufacturer’s instructions, including mechanical cleaning (e.g. brushing, wiping). When using in ultrasonic bath, cover the bath during sonication. Rinse with water of at least drinking quality. If disinfection is done subsequently, rough drying of the outer surfaces and particularly of cavities e.g. endoscope channels, is necessary in order to rule out any dilution of the disinfectant solution.
For the further terminal disinfection step, we recommend our product: Sekusept aktiv
Proteases, amylase, lipase, surfactants, corrosion inhibitors, solvents, additives, emulsifiers, perfume.
|Use (after commencement)||Shelf life|
Please note the following to ensure the correct storage of environmental hygiene and instrument disinfectants: The original containers (canisters, buckets or bottles) with the disinfectant concentrate (or powder) should be stored at temperatures below 25°C and kept dry and clean. A brief period below or above the given storage temperature (e.g. during transport) will not diminish the effectiveness.
The containers are to be sealed tightly after each use. Please also take care that the concentrate remaining in the container is not contaminated when preparing the application solution.
Assuming that the application solution that is produced is not contaminated by organic material, it is to be stored in a clean, closed container or a covered instrument basin.
Please take note of the corresponding marking on the containers in which the disinfectant solutions are kept, e.g. contents, concentration, date, shelf life and batch number. Used and organically contaminated solutions are to be changed at least once per workday as per RKI recommendation.
|Chemical- physical data|
|Density||20 °C||1.06 g / cm3|
|pH value* (concentrate)||20 °C||approx. 8.6|
|pH-range * (1-2% solution)||20 °C||approx. 6 – 7|
* pH-value measurements under operating conditions always using pH-meter with temperature compensation.
Cleaning of medical devices